If you’re having problems with the water pump within your Tuttnauer autoclave you have to discover what the issue is right away and fix it. No water or even a water level that’s as well low indicates your machine isn’t working properly and won’t be completely sterilizing your equipment. And that means you’re basically from company before you get the issue solved.
If you’ve constantly sent your autoclave out for repair anytime there’s a difficulty then you know just how expensive these fixes can be. You have labor charges, costs for components, and don’t ignore the handling and shipping charges. You might nearly purchase a new machine when all is considered and done! If you’re having problems with the water within your Tuttenauer autoclave, below are a few problem solving tips that might assist you to resolve the issue yourself.
When the water is moving in the incorrect path, check the 2 tubes attached to the water pump. It may be that you just must reverse them.
In the event the pump is creating sound check to see if the installation bracket is loose or broken. You may just must tighten up it down or change it. If it’s not the bracket the problem is most likely within the water pump itself and you’ll need a new one.
In the event the water pump keeps coming a fuse, first search for pinched or twisted wires that might be creating a brief. If that’s not the situation, turn the water pump away and disconnect the wires through the number 1 and 2 terminals on the solid state relay. Transform the water pump back on and when the fuse blows again the issue is with either the water water pump or even the pump capacitor and you’ll need to replace both.
When the pump won’t turn off you will want a voltage/ohmmeter to check the relay. Read among TP13 and TP1. A 10 to 12 volt DC transmission must tell the communicate to change away along with a to 1 transmission is necessary to inform it to turn on. When the reading is other things it indicates you do have a downside to the control circuit.
In the event the signal is proper, repeat the method on the relay. And when you receive an wrong transmission there then you’ll need to change the Ajunc board.
In case your pump won’t switch on, the first thing you have to do is check the fuse and replace it if needed. Following, look for pinched or twisted cables leading from the strong state relay for the water pump. If ejxawh doesn’t help, transform the power to your autoclave away and disconnect the cables on terminals 1 and 2 around the relay. Now, link the two cables together and transform the water pump back on. When the pump nevertheless doesn’t work then you’ll must replace it.
Whilst your autoclave might be a ‘machine’ and it performs a very important, life-conserving function, it’s still, in the end, just a huge box with some basic operating components. It’s generally easy sufficient to obtain the issue. And more often than not you can remedy it your self!